- Em promoção!
Grove - Kit sensor de PH (E-201C-Blue)
Este produto é um sensor/medidor de pH, de baixo custo e fácil utilização, adequado para Arduino e Raspberry Pi, suporta sistema de 3,3 V e 5 V.
Se tiver alguma dúvida neste produto não hesite em contactar-nos.
*Atenção: as imagens são meramente ilustrativas.
- Widely used in chemical industry, pharmaceutical industry, dye industry, and scientific research
- Support with both Arduino and Rasberry Pi
- Compact size for easy deployment and cost-effective
- Resolution: at most ±0.15PH (STP)
- Probe replaceable
Are you trying to find an easy to use and cost-effect PH sensor/meter? Do you want to use a PH sensor/meter with Arduino or Raspberry Pi? Well, this new Grove - PH Sensor will meet all your needs. The PH sensor measures the hydrogen-ion activity in water-based solutions, we usually use it to measure the PH of a liquid. It is widely used in the chemical industry, the pharmaceutical industry, the dye industry, and scientific research where acidity and alkalinity testing is required. The drive board in this kit support both 3.3V and 5V system. And with the stander BNC probe interface and Grove connector, it is very convenient to work with Arduino and Raspberry Pi.
This product is non-RoHS certified.
- Before being measured, the electrode must be calibrated with a standard buffer solution of known PH value. In order to obtain more accurate results, the known PH value should be reliable, and closer to the measured one.
- When the measurement is completed, the electrode protective sleeve should be put on. A small amount of 3.3mol / L potassium chloride solution should be placed in the protective sleeve to keep the electrode bulb wet.
- The leading end of the electrode must be kept clean and dry to absolutely prevent short circuits at both ends of the output, otherwise it will lead to inaccurate or invalid measurement results.
- After long-term use of the electrode, if you find that the gradient is slightly inaccurate, you can soak the lower end of the electrode in 4% HF (hydrofluoric acid) for 3-5 seconds, wash it with distilled water, and then soak in potassium chloride solution to make it new.
|Saturated sodium chloride solution||1|
What is Grove?
Grove makes it easier to connect, experiment, and simplify the prototyping process. No jumpers or soldering required. We have developed more than 300 Grove modules, covering a wide range of applications that can fulfil a variety of needs. Not only are these open hardware, but we also have open-source software.
For all Grove users (especially beginners), we provide you with the guidance of operation. Please read the instructions through the official website before your using the product.
LEARN AND DOCUMENTS
[Wiki] PH Sensor Kit (E-201C-Blue )
This is the wiki page for this product, which will show you how to use the product, as well as details about the software and hardware.
Biofloc fish project monitoring. Water quality maintenance and monitoring in any aquaculture are the essential practices aiming at the success of the growing cycles.
Water Quality is a complex concept, it related to too many parameters. Therefore constructing a water quality monitoring station is a systematic and complex project. The good news is that with the help of those 5 Grove Water Quality Sensors, you can build your own lightweight water quality monitoring station easily.
Também poderá gostar